Force decay of intermaxillary orthodontic elastics: in vitro study
AbstractIntroduction. Orthodontic elastics are widely used in orthodontic treatment. It’s been proved that they have advantages such as low cost, ease of use, but also disadvantages, mainly force decay in time and increased entrapment of biofilm. Amount of the force is of extreme significance. This force can be altered by physical or chemical factors. Material and Methods. Latex elastics in 3 different diameters were selected for this study. Each elastic was stretched and placed on hooks that are at specified distances that equal 3 times the diameter of each elastic. The forces produced by stretching was measured using tension gauge and the measurement were taken at specific time intervals of 0h, 3h, 12h, 24h. The same process was repeated for elastics in dry and artificial environment. Results. Elastics in the dry environment showed progressive force decay cause by stretching over time. Just after 3 hours force decay between 6,07% and 8,75% was observed. The biggest loss of force between 13,61 – 16,13 % was measured after 24 hours. Compared to the dry environment, an even more significant force decay was observed in the artificial saliva. After first 3 hours force loss was between 4,99–9,22%. The biggest force decay was observed after 24 hours and it was 5 % higher compared to dry environment. Conclusions. 1.The artificial saliva environment and time of exposure to it, have a negative effect on the properties of elastomeric. 2.To maintain the effective orthodontic strength of elastics, they should be replaced every 12 hours.
Lacerda Dos Santos R., Pithon M.M., Romanos M.T., The influence of pH levels on mechanical and biological properties of nonlatex and latex elastics. The Angle Orthodontist, 2012, 82, 709-714.
Stoner MM. Extraction treatment. In: Graber TM, editor. Orthodontic concepts and techniques, volume 1. Philadelphia: W. B. Sa- unders; 1969.
Andreasen GF, Bishara SE. Comparison of alastik chains with elastics involved with intra-arch molar to molar forces. Angle Orthod ?1970;40:151-8. ?
BisharaSE,AndreasenGF. A comparison of time related forces be?tween plastic alastiks and latex elastics. Angle Orthod 1970;40: 319-28. ?
Buchmann N, Senn C, Ball J, Brauchli L. Influence of initial strain on the force decay of currently available elastic chains over time. Angle Orthod 2012;82:529-35. ?
Ash JL, Nikolai RJ. Relaxation of orthodontic elastomeric chains and modules in vitro and in vivo. J Dent Res 1978;57:685-90. ?
de Souza RA, de Araujo Magnani MB, Nouer DF, da Silva CO, Klein MI, Sallum EA, et al. Periodontal and microbiologic evalu- ation of 2 methods of a archwire ligation: Ligature wires and elas- tomeric rings. Am J Orthod Dentofacial Orthop 2008;134: 506-12.
Sauget, P.S., Stewart K.T., Katona, T.R., The effect of pH levels on nonlatex vs latex interarch elastics. The Angle Orthodontist, 2011, 81, 1070-1074
Beattie S., Monaghan P., An in vitro study simulating effects of daily diet and patient elastic band change compliance on orthodontic latex elastics. The Angle Orthodontist, 2004, 74, 234-239
Pithon M.M., Souza R.F., Freitas L.M.A. i in., Mechanical properties intermaxillary latex and latex-free elastics. Journal of the World Federation of Orthodontists, 2013, 2, 15-18
Loriato L.B., Machado A.W., Pacheco W., Clinical and biomechanical aspects of elastics in Orthodontics., Dental Press Journal of Orthodontics, 2006, 5, 43-55.
Fernandes D.J., Fernandes G.M., Artese F., i in., Force extension relaxation of medium force orthodontic latex elastics. The Angle Orthodontist,2011, 81, 812-819
Loch J. , Krawiec H., Zachowanie korozyjne stopów kobaltu w roztworze sztucznej śliny, Archives of Foundry Engineering, 2013, 13, 101-106
Kumar K, Shetty S, Krithika MJ, et al., Effect of commonly used beverage, soft drink, and mouthwash on force delivered by elastomeric chain: a comparative in vitro study, Journal of International Oral Health, 2014, 3, 7-10.
Pithon MM, Mendes JL, da Silva CA, i in., Force decay of latex and non-latex intermaxillary elastics: a clinical study, European Journal of Orthodontics, 2016, 1, 39-43